ATS Techniques and Tips Selected ATS Journal articles useful for restoring,
cleaning maintaining telescopes, books, etc.
A Professional Method
for Cleaning Optics
by Robert Ariail
- (From the Journal of the ATS - 1995, Issue #8)
The proper cleaning of optical surfaces, while not in itself a restorative process, is an integral part of
the proper care and function of any optical instrument - antique or modern. Furthermore, every
telescope owner should be aware of the process that I am about to describe.
There are many avowed cleaning techniques which in fact work very well and have been used
throughout the years by professionals and knowledgeable amateurs as well. It is also a reality that
all cleaning methods and techniques, no matter how refined or skillfully applied, result in a
minute amount of damage to the delicate optical surface each time it is cleaned. All
techniques, that is, except one. This single exception is in many ways the best method of all. It
involves the simple application of a chemical substance to the lens or mirror on hand. Its use
requires little skill, eliminates completely the rubbing or touching of the surfaces, and thoroughly
removes all grease, dust and dirt with a single application. The final result is an almost antiseptic
surface. This magical cleaner of mirrors and lenses is Collodion.
This technique was apparently first reported by James B, McDaniel in 1964. He stated: "Perhaps
the most complete cleaning of an optical surface short of doing an ion glow discharge cleaning in a
vacuum bell jar is achieved by using the Collodion technique of mirror cleaning. Surfaces cleaned
by this method have perhaps 10 times less residual contaminants and particulates as
compared to a methanol-distilled water cleaned surface."
In 1970, John B. Tyndall produced a concise report with improvements on this procedure.
Essentially, Tyndall introduced the addition of a layer of cheesecloth embedded in the
Collodion. This aided in peeling the dry Collodion from the mirror. He recommended applying a
series of thin coats with a camel's hair brush.
At the suggestion of Dr. Donald E. Brownlee of the University of Washington Astronomy
Department, Edward J. Mannery (Manastash Ridge Observatory, University of Washington) tried
the technique on several small mirrors with beautiful results. Following that success, he dared to try
it on the University of Washington 30" telescope which needed cleaning once a year. Again,
the results were better than could have been obtained by any other method.
My personal experiences have been similar. On small to mid-sized mirrors, Collodion is a dream
to use. Another excellent advantage not yet mentioned is the fact that the mirror can be cleaned in
its holder without the need of removing it. Objective lenses can likewise be cleaned while retained
in their cells. I am of the opinion that Collodion adheres more efficiently to the aluminum of the
mirror than to the glass of the objective lens. That is not to say that the lens can not be cleaned as
effectively, but a modicum of finesse may be required. In the case of the lens, the Collodion tends
to part and leave small residual patches on the surface. These can be removed with a strip of
masking or scotch tape gingerly touched to the errant patches.
Certainly one of the largest lenses ever cleaned with Collodion has to be the great 23-inch Alvan
Clark objective now at the Charles E. Daniel Observatory, Greenville, S.C. After it was taken out
of the container where it had been stored for many years, it was in dire need of cleaning. The R1
surface of the crown was especially dirty. I suggested to Doug Gegen, Astronomer of the
observatory, that we attempt to clean it with Collodion, He agreed, and we went to work. I can still
vividly remember how apprehensive I was as we poured the sticky substance over the giant lens!
My fears abated when the Collodion pealed off in even sheets ... save a few patches ... leaving a
magnificently clean surface. Incidentally, the remaining surfaces had to be cleaned while in the
cell conventionally. The 7-inch air spacing between crown and flint allowed for such procedure.
Dick Nelson, founder of The Optical Craftsman telescope company and engineer for the
observatory project accomplished this.
In using Collodion, simply remember the following guidelines:
1. Use only U.S.P. Collodion (cellulose nitrate in ether-methanol solution). It may be obtained
from any drug store or pharmacy. Flexible Collodion should not be used.
2. Apply the Collodion by simply pouring, spraying, or laying it on with a camel's hair brush.
3. If the mirror or lens is in its cell, it may be necessary to make a dam of masking tape,
cardboard or other material to prevent the collodion from seeping between glass and cell. If applied
with a brush in thin coats, the dam may not be necessary.
4. Adding a layer of cheesecloth or surgical gauze while the Collodion is still wet will make it
easier and simpler to peel the substance off in a single sheet when dried. For small optics, the gauze
may not be necessary or desirable.
5. After application, let the Collodion dry before removal. Generally, telltale signs that it is ready for
removal are slight shrinkage, and curling or lifting of the leading edges. At this stage, simply (slowly
and carefully) peel it off in a single sheet, if possible. If patches remain, mop up carefully with
6. Work in a well ventilated area (Collodion is ether-based). Also, gloves may be desirable for
7. Enjoy the cleanest optics you have ever seen!
(Note from Glenn Oclassen: You will probably need to have your druggist order your
collodion supply from the manufacturer. Don't let him sell you flexible collodion, only plain
Collodion USP. A 500 ml bottle cost me about $30 in July 1995)